Why Did My Pressure Equipment Fail? 6 Instrument Killers.Your process pressure instruments play a critical role in your applications. They accurately and reliably measure and monitor pressure to determine if process systems are working effectively and efficiently, or if you need to make system alterations or adjustments. This helps ensure the safety of your personnel, the efficiency of your processes and the quality of your products.
But what happens when your pressure instruments break down?In my 41 years of working in the process industry, I’ve seen many causes of instrument failure. These “instrument killers” can throw a wrench in your project and cause all types of problems. Luckily, most of them can be prevented if you know the warning signs. In this article, I will outline the most common instrument killers that endanger your pressure equipment. Although the focus of this blog is on pressure gauges, when the word instrument is used, the reference is to a gauge, a pressure switch, or a transducer. For more information on pressure gauge standards, refer to ASME B40.100-2013 and EN837-1. I’ll also teach you how to see these instrument killers coming ahead of time and how to solve them so you can keep your processes running smoothly, safely and accurately.
Pressure Instrument Problem Warning SignsInstrument maintenance is one of the keys to achieving and maintaining personnel safety, process efficiency and product quality. A comprehensive maintenance plan makes it easier for you to identify instrument issues and execute appropriate solutions before they become more significant or widespread problems. This helps save you time and money by preventing costly service work, unplanned downtime and equipment or process-based employee injuries.
Early warning signs of gauge issues include:Pointer Flutter
The gauge pointer continually moves back and forth, making it difficult to read the measurement indicated.
The pointer component does not move in response to changes in the operating pressure for which it is designed.
Various components of the gauge may have damage. For example, the pointer may be dented, a sign the gauge has been severely under or overpressure. In a liquid-filled gauge case, yellowed fill usually indicates elevated ambient temperature at the gauge. Also, the dial may be discolored, which is another sign of elevated temperature.
A melted or deformed gauge window, usually acrylic, is also a sign of extreme elevated temperature. The gauge window and dial are coated in a black powder making gauge readability an issue. This is typically the result of excessive pulsation and/or vibration that causes fretting or wear of the gauge movement teeth. Among others, these signs serve as indicators of underlying gauge issues. If you can recognize them and understand their causes, you will be better prepared to diagnose and resolve problems that may occur in your own process instruments.
Common Pressure Instrument Problems and SolutionsLike many of my colleagues in the industry, you may also encounter problems when using instruments in your critical applications. The key to overcoming them is recognizing their signs and understanding their causes. This information makes it much easier to determine and execute the right solution. Below are some of the most common pressure instrument problems, including why they occur, what effects they have on the instrument and the overall operation or process, and how to resolve them.
ProblemIn processing equipment, surges in process pressure can lead to pulsation in gauges. These pulsations can cause poor gauge readability due to point flutter and decreased gauge integrity due to component damage. Similar to pulsation, vibration can lead to pointer flutter and component damage in gauges. They can be classified into two categories: high-frequency/low-amplitude and low-frequency/high-amplitude. The former causes pointer flutter and centralized gauge movement gear teeth damage on the segment, while the latter causes widespread gear teeth damage across the segment and increased stress on the Bourdon tube. As a rule of thumb, if pulsations or vibrations trigger pointer movement of 5% or more of the full-scale range, you should take steps to reduce their generation or their effect on the instrument.
SolutionThere are three common solutions for instrument problems stemming from excessive pulsations or vibrations:
- Dampening the movement/tube
- Installing the instrument assembly away from the source
- Restricting the flow of the process material
ProblemInstruments are rated for use in a specific temperature range. Using them in temperatures outside of this range can lead to significant damage, depending on their design and construction. For example: Dry gauges are typically rated for use in ambient temperatures below 200°F (93°C). Above this temperature, they may experience dial discoloration, window discoloration (if acrylic) or gasket hardening. Process temperatures for dry gauges are typically rated for between -20°F (-20°C) to 250°F (121°C). Liquid-filled gauges are typically rated for use in ambient temperatures up to 150°F (66°C). Above this temperature on a liquid-filled gauge they may experience leaching of the gasket and O-ring material and discoloring of the fill liquid. Both extremely low and extremely high temperatures can also trigger accuracy issues in instruments. A standard dry gauge can experience slowed point response time in low temperatures. Gauge accuracy is affected when ambient gauge temperature exceeds the temperature at which it was calibrated. As a guide, if a gauge is calibrated at 68°F (20°C), accuracy will be affected by approximately 0.4% per 25°F/13.9°C.
SolutionThere are several common solutions for instrument problems stemming from excessively high or excessively low temperatures:
- Mounting the instrument remotely
- Mounting the instrument directly
- Incorporating a coil or pigtail siphon for steam applications
Problem:In my time working in this industry, I’ve learned that pressure spikes have many causes, including water hammer, overly rapid valve actuation, equipment malfunction or process fluid freezing. Gauges subjected to these conditions often show signs of damage like a dented pointer, a ruptured or deformed tube, or a broken segment gear. This damage can lead to unresponsiveness to changes in process conditions or complete instrument failure, both of which can result in poor operation or performance in your process. Pressure spikes or surges beyond the full-scale range of a gauge or overpressure can result in accuracy degradation, gauge failure or worst-case rupture of the bourdon tube.
Solution:There are many solutions if you experience instrument problems stemming from pressure spikes, including:
- Integrating an internal stop
- Installing a pressure limiting valve (PLV) or a gauge with high overpressure capability
- Selecting a new, properly rated gauge
ProblemInstruments installed in equipment for “dirty” processes—i.e., processes involving particulates, slurries and sludge—are highly susceptible to clogging. Process material can become stuck on the instrument’s inner surfaces as it flows through the system, which can hinder the operation of the instrument and the efficiency of the process.
SolutionIf the instrument’s process connection is susceptible to process media buildup or blockage, you can prevent clogging by isolating the instrument from the process media. A diaphragm seal or isolation ring are good choices. When clogging of the diaphragm seal is a concern, consider an available flushing connection. It is designed to flush process media buildup in the seal. Attaching a valve to the flushing port is ideal to control process flow. For heavy slurries or sludge, consider an isolation ring. This 360-degree design for process flow makes clogging virtually impossible.
ProblemThe wetted parts of an instrument must be compatible with the process material. Otherwise, the instrument may corrode during use, which can affect product and process quality. Under severe process conditions, process media that is not compatible with the wetted parts of the instrument will result in eventual failure of the instrument causing the media to escape into the environment. This becomes a safety issue where operators can be injured or, in extremes, lose their lives.
SolutionIf you want to avoid instrument problems stemming from corrosion, you should ensure the material for the instrument’s wetted parts is suitable for the process material’s composition, concentration and temperature. If it is not possible to choose a compatible instrument wetted parts material, you can integrate a diaphragm seal/instrument isolator or isolation ring constructed from an appropriate material for the process. Refer to the Ashcroft corrosion guide for process compatibility guidance.
ProblemInstruments are delicate pieces of equipment. While integrating the right design elements and accessories can help protect them from damage during severe service applications, it cannot protect them against abuse, i.e., incorrect installation or usage. Installing or using instruments incorrectly can cause significant damage, which can lead to measurement error or component failure.
SolutionIf you want to avoid instrument problems stemming from instrument abuse, you should install, use and maintain your instruments properly. Installing NPT instruments requires the use of an open-ended wrench. Do not attempt to install and tighten the instrument using the instrument case! This often causes irreparable damage to the instrument. Do not use the instrument as a step ladder after it is installed as this presents safety issues. Not only can the instrument be damaged, but someone can get hurt. Inspect the instruments routinely for damage and performance.
We don’t like to pressure you, but we have more information.I hope you are now equipped with the knowledge you need to keep your people, processes and profits safe and secure. If you know what problems to look out for and how to solve them, you won’t have to fear “instrument killers.” If you want to learn more about pressure instruments, feel free to visit our website and view any of our helpful product documentation, white papers, webinars or guides. At Ashcroft, we are committed to helping customers select and source the right instruments for their needs. Whether you need assistance choosing instruments for a new process or operation, or resolving instrument issues with an existing equipment setup, we’ve got you covered. Contact us today to talk to one of our industry experts and get all your questions answered.
About the Author
Global Product Manager
Lou Altieri is a Global Product Leader at Ashcroft Inc. with more than 41 years of experience. He is responsible for pressure gauges, diaphragm seals, isolation rings and accessories for the oil and gas, chemical/petrochemical and water/wastewater markets. Lou has a passion for understanding customers' needs and providing solutions to their problems. Recently, to safely protect the gauge from elevated pressure beyond full scale, he launched a gauge that can withstand pressure up to 4 times full scale without damaging the gauge. He has authored numerous articles. In his spare time, Lou enjoys power walking, hiking and viticulture.